Jugaad PCR Thermocyclers
Coconut PCR :-)
some discussion about the light bulb and fan thermocycler in DIYbio maillits.
referring to an indian research group
cathal harvey showed me the hot airgun version with relay, no posts found yet on indie biotech
open hardware thermocycler
The classic OpenPCR kit to order, assemble and hack by openPCR.com
OpenPCR in the wild
Attempt to build an OpenPCR just from the desctription on the website OpenPCR without a kit or any supplied parts.
First and easy thing is to go up to a FabLab and laser cut the wood parts for the housing. Need to get the files from the website and have the most recent version of SolidWorks to extract the drawings...
Here we go;
And the one pdf with all the parts as of November 2012:
Got some additional information from Josh from OpenPCR: The downloadable files are the more recent versions of our design.
The OpenPCR BOM can be found here:
Good to know.
What else do we need?
- The PC-Cooler with 90° heat pipe is easy to get for around 20 euro.
- The nice and small power supply that fits into the housing is not so easy. It is a "Flex 1u PC Power Supply 200 W ATX. Bigger (standard size is more easy and should do, just won't fit into the housing
- Arduino ok, SMD driver board with AD converter, connectors, feasible. Will try to simplify the layout to make it single-side DIY. What are the big resistors for? Some kind of heating? Have to figure out.
- LCD Display - standard, anything goes, 4 Lines are nice.
Now one tricky part is the heatblock and the peltier element. For those we have two nice hacks: 1. A peltier element with the exact right dimensions and with some luck even the rough thermo-block can be found in one of those little electric fridges:
Found those in electronic waste.
Unscrew the blower and the cooler and you will find the peltier element.
All you need, a aluminium heat-block (think the original OpenPCR heat-block is cooper and then chromed, right?), peltier element...
Now that we have the heating/cooling element and the block we need to get the right holes for the eppendorf tubes in it.
Josh says from his experience: "Different manufacturers make 200 ul tubes of different dimensions. But we designed OpenPCR around the key angle of 17.5 degree cone angle, which seems to work for most PCR tubes. Good luck in your endeavors".
So with a fried we designed a special milling tool around this angle. We make this an "Open Shared Hardware Tool", so who ever needs to make some nice 17.5 degree holes may borrow these for a while :-)
This is the drawing with dimensions: