HiSeq2000 - Tear Down



The overall look of HiSeq2000:
 * 1 For the right side view,
 * 2 For the front view of the syringes(A) and the fridge(B),
 * 3 For the left side view,
 * 4 For the front view of the optical bench(C).

----

Sockets
 * 1A: on Motherboard; 1B: Fan; 2A: Syringe pump; 2B: Fridge for reagent:


 * 3A: the other side of Motherboard; 3C: two camera control board; 4: front view of the stage


 * Wires: 1 (Camera & Frame Grabber | PE Prime)




 * Wires: 3 (MainBoard Socket | MainBoard Wires)





(Parker_XY_Stage Board & Wires):



(Laser & Camera | Docking Stage & Sensors)






 * Vacuum Pump

Red tubes:

Pisco Polyurethane tubing 3.18x2 * UB-1/8-R * 86-KT121;

UB: http://www.pisco.com/products/Polyurethane_Tubing

Polyurethane tubing->UBT 1/8 Red: http://www.pisco.com/OrderTubing

Connectors:

Datasheets from SMC: File:KQ2-Dd_EU.pdf

https://www.smc.eu/portal_ssl/WebContent/digital_catalog_2/jsp/view_subclasses.jsp?dc_product_id=159156#s

Behind the optical bench (between red tubes and green tubes):

SMC One-touch Fittings: Different diameter straight: KQ2H:

Ø 3.2-Ø 4: KQ2H23-04A1:

https://www.smc.eu/portal_ssl/WebContent/digital_catalog_2/jsp/view_product_configurator.jsp?dc_product_id=159183

Between the green tubes and the filter:

SMC One-touch Fittings: Union “Y”: KQ2U:

Ø 4: KQ2U04-00A:

https://www.smc.eu/portal_ssl/WebContent/digital_catalog_2/jsp/view_product_configurator.jsp?dc_product_id=159189

SMC One-touch Fittings: Plug: KQ2P:

Ø 4: KQ2P-04:

https://www.smc.eu/portal_ssl/WebContent/digital_catalog_2/jsp/view_product_configurator.jsp?dc_product_id=159217

On the Filter:

SMC One-touch Fittings: Branch "Y": KQ2U (Sealant):

Ø 3.2: KQ2U23-01�S/ KQ2U23-02�S (Since Connection thread NOT SURE):

https://www.smc.eu/portal_ssl/WebContent/digital_catalog_2/jsp/view_product_configurator.jsp?dc_product_id=159176




 * Cooling system/Coolant

Hoses, Fans, Sensors and Coolant: Koolance https://koolance.com/products

Hoses: Tubing: ID: 10mm (3/8in): https://koolance.com/index.php?route=product/category&path=94_96

Under Optical bench: PLT-UN50F Cold Plate, 50mm x 50mm (1.97in x 1.97in): Part No. PLT-UN50F

https://koolance.com/cold-plate-50mm-plt-un50f

Flow Sensors: INS-FM16 Coolant Flow Meter: Part No. INS-FM16

https://koolance.com/ins-fm16-coolant-flow-meter

Fans:

Fans: Fan, 120x25mm, 108CFM(Part No. FAN-12025HBK) OR Fan, 120x25mm, 90CFM(Part No. FAN-12025MBK) (NOT SURE)

https://koolance.com/index.php?route=product/category&path=83

Fan Grill, 120mm: Part No. GRL-AP002P:

https://koolance.com/fan-grill-120mm

Quick Connectors:

VL2N Male Quick Disconnect No-Spill Coupling, Threaded G 1/4 BSPP: Part No. VL2N-MG

https://koolance.com/vl2n-quick-disconnect-no-spill-coupling-male-threaded-g-1-4

VL2N Female Quick Disconnect No-Spill Coupling, Threaded G 1/4 BSPP: SKU: VL2N-FG

https://koolance.com/vl2n-quick-disconnect-no-spill-coupling-female-threaded-g-1-4

Spring Warps: Tubing Spring Wrap, Steel Black for OD 13mm (1/2in): Part No. SPR-10BK

https://koolance.com/tubing-spring-wrap-steel-black-for-od-13mm-1-2in

Coolant: Koolance 702 Liquid Coolant, High-Performance, UV Yellow, 700ml (24 fl oz): Part No. LIQ-702YL-B

https://koolance.com/liq-702-liquid-coolant-bottle-high-performance-700ml-uv-yellow




 * Cooling System Board/Coolant Board




 * Reagnet Chiller Board



Full Version PDF:

03.Dec.2018: File:Illumina HiSeq2000 TearDown 03.12.2018.pdf

15.Feb.2019:  File:Illumina HiSeq2000 TearDown 15.02.2019.pdf